Development and evaluation of an audiology app for iPhone/iPad mobile devices
Larrosa F (1), Rama-Lopez J (2), Benitez J (3), Morales JM (4), Martinez A (4), Alañon MA (4), Arancibia-Tagle D (2), Batuecas-Caletrio A (5), Martinez-Lopez M (6), Perez-Fernandez N (6), Gimeno C (7), Ispizua A (8), Urrutikoetxea A (9), Rey-Martinez J (10).
(1) Department of Otolaryngology, Hospital Alt Penedes , Barcelona, Spain.
(2) Department of Otolaryngology, Hospital Universitario Son Espases , Palma de Mallorca, Spain.
(3) Department of Otolaryngology, Hospital Universitario de Gran Canaria Dr. Negrín , Las Palmas, Spain.
(4) Department of Otolaryngology, Hospital General Universitario Ciudad Real , Ciudad Real, Spain.
(5) Department of Otolaryngology, Hospital Universitario de Salamanca, IBSAL , Salamanca, Spain.
()6 Department of Otolaryngology, Clinica Universidad de Navarra , Pamplona, Spain.
(7) Department of Otolaryngology, Clinica Rotger, Palma de Mallorca , Mallorca, Spain.
(8) Department of Otolaryngology, Hospital Universitario Donostia , San Sebastian, Spain.
(9) Department of Otolaryngology, Hospital Comarcal del Bidasoa , Hondarribia, Spain.
(10) Otolaryngology Unit ORLGipuzkoa, Clínica Quirón Donostia , San Sebastian, Spain.
Magazine: Acta Oto-laryngologica
Date: Oct 1, 2015Otorhinolaryngology
The application described in this study appears to be accurate and valid, thus allowing calculation of a hearing handicap and assessment of the pure-tone air conduction threshold with iPhone/iPad devices.
To develop and evaluate a newly developed professional, computer-based hearing handicap calculator and a manual hearing sensitivity assessment test for the iPhone and iPad (AudCal).
Multi-center prospective non-randomized validation study. One hundred and ten consecutive adult participants underwent two hearing evaluations, a standard audiometry and a pure-tone air conduction test using AudCal with an iOS device. The hearing handicap calculation accuracy was evaluated comparing AudCal vs a web-based calculator.
Hearing loss was found in 83 and 84 out of 220 standard audiometries and AudCal hearing tests (Cohen's Kappa = 0.89). The mean difference between AudCal and standard audiogram thresholds was -0.21 ± 6.38 dB HL.
Excellent reliability and concordance between standard audiometry and the application's hearing loss assessment test were obtained (Cronbach's alpha = 0.96; intra-class correlation coefficient = 0.93). AudCal vs a web-based calculator were perfectly correlated (Pearson's r = 1).
CITATION Acta Otolaryngol. 2015;135(11):1119-27. doi: 10.3109/00016489.2015.1063786. Epub 2015 Jul 4.
you mayBE INTERESTED
The Clínica is the greater private hospital with technological equipment of Spain, all in a single center.
The professionals of the Clínica perform continuous research and training, always to the benefit of the patient.
Learn why we are different from other healthcare centers. Quality, speed, comfort and results.